When using the optional mass spectrometer, the magnet is energized by a Electronic Measurements 600V supply. In order to achieve sufficient precision in setting the magnet current it is best to control the EM supply with the motorized potentiometer. Both pieces are located in the SCAM rack.
Cross Vacuum Chamber
This page is maintained by Bill Russell.